Electronic Components Datasheet Search
K-TS1 Datasheet(PDF) 5 Page - RFbeam Microwave GmbH
RFBEAM [RFbeam Microwave GmbH]
K-TS1 RADAR TESTSYSTEM
Startup Message and Sections
K-TS1 sends the following screen after receiving an [Enter] key from the terminal program:
RFbeam K-TS1 24GHz Test System #09160219
Program Version V3.01 Jan 27 2011
[c] CW Carrier on/off
[t] Target on/off
24.125 [23.000 .. 25.000GHz]
[q,w,e] Frequency ++
[i,o,p] Frequency â€“
[g] Target Frequency
[-9999 .. 9999Hz]
[r] Store to EEPROM
[b] Jump in Bootloader
[m] Measure f/P once
[l] Measure f/P in loop
[a] Cyclic table update
[-] Transmitter EIRP
[-] IF1 power
[-] IF2 power
[-] AUX power
[-] Target EIRP
K-TS1 measuring results
displayed at command "a1"
K-TS1 command and result line
Fig. 4: K-TS1 output structure at a standard terminal program
K-TS1 Command Set
All commands must be followed by an [Enter] key (ASCII 0x0d). Exception: Fast up ([q], [w], [e]) and
down ([i], [o], [p]) adjustment keys. Cyclic operations can be cancelled by [Enter] or [Esc] keys.
0 | 1
K-TS1 transmits a CW carrier at a frequency set by [f].
0 | 1
Turns on the Doppler target simulator. The signal received at "target Rx antenna" will be amplified, SSB
modulated by the programmable target frequency and sent back via the "target Tx antenna". Tip: do not
use cyclic inline display [l] when using this function. Otherwise there will possibly be distortions in the
Carrier frequency is set to f with a resolution of 1 MHz.
Please note, that all digits including the decimal point must be entered.
Fast frequency increase by 1MHz, 10MHz, 100MHz. This is very handy for frequency sweep by a host
i, o, p
Fast frequency decrease by 1MHz, 10MHz, 100MHz
Target simulator frequency may be set with a resolution of 1Hz. The sign sets the simulated direction of the
target. A positive value will simulate a forward moving target and a negative value a backward moving
target. Please note, that all digits including leading zeroes must be entered.
Range for K-TS1 with Firmware smaller then V3.01 = 0 .. 4000Hz
Range for K-TS1 with Firmware equal or higher then V3.01 = -9999 .. 9999Hz
Calibration and special mode settings values for the K-TS1 system. See Special I/O Settings
Nonvolatile storage of the actual settings. Use this functions for standalone operation of K-TS1
This command is used for updating K-TS1 firmware. Refer to the release notes on http://www.rfbeam.ch/
Performs a "oneshot" measurement of Rx frequency, Rx power and Power at the "Aux" connector
Performs a "cyclic" measurement of Rx frequency, Rx power and Power at the "Aux" connector. Stop cyclic
display by [Esc]. Tip: Switch doppler target off when using this function.
0 | 1
Performs a "cyclic" measurement of internal values (refer to Fig. 4). Turn CW carrier on to get values.
Enters commands and refreshes the screen
Escapes/stops running command [l] and refreshes the screen
Â© RFbeam Microwave GmbH www.rfbeam.ch
Does ALLDATASHEET help your business so far?
[ DONATE ]
All Rights ReservedÂ©
|   Chinese :
| German :
| Japanese :
| Russian :
| Spanish :
| French :
| Italian :
| Portuguese :
| Polish :